This document specifies a method for the determination of nitrate as NO3-N in water of various origin such as natural water (including groundwater, surface water and bathing water), drinking water and wastewater, in a measuring range of concentration between 0,10 mg/l and 225 mg/l of N03-N using the small-scale sealed tube method. Different measuring ranges of small-scale sealed tube methods can be required.
The measuring ranges can vary depending on the type of the small-scale sealed tube method of different manufacturers.
It is up to the user to choose the small-scale sealed tube test with the appropriate application range or to adapt samples with concentrations exceeding the measuring range of a test by preliminary dilution.
NOTE 1 The results of a sealed-tube test are most precise in the middle of the application range of the test.
Manufacturers' small-scale sealed tube methods are based on dimethylphenol colour reaction depending on the typical operating procedure of the small-scale sealed tube used, see .
NOTE 2 Laws, regulations or standards can require that the data is expressed as NO3- after conversion with the stoichiometric conversion factor 4,426 81 in .
NOTE 3 In the habitual language, use of sewage treatment and on the displays of automated sealed-tube test devices, NO3 without indication of the negative charge has become the common notation for the parameter nitrate and especially for the parameter nitrate-N. This notation is adopted in this document even though not being quite correct chemical nomenclature.
État actuel: PubliéeDate de publication: 2023-02
Comité technique: ISO/TC 147/SC 2 Méthodes physiques, chimiques et biochimiques
Acheter cette norme
|std 1 61||PDF + ePub|
|std 2 61||Papier|